|Year : 2019 | Volume
| Issue : 4 | Page : 305-310
Laparoscopic and robotic specimen retrieval system (Modified Nadiad Bag): Validation and cost-effectiveness study model
Chaitanya Sharad Deshmukh1, Arvind P Ganpule1, Mohammad Rafiqul Islam2, Ravindra B Sabnis1, Mahesh R Desai1
1 Department of Urology, Muljibhai Patel Urological Hospital, Nadiad, Gujarat, India
2 Department of Urology, Bangabandhu Sheikh Mujib Medical University, Dhaka, Bangladesh
|Date of Submission||19-May-2018|
|Date of Acceptance||23-May-2018|
|Date of Web Publication||10-Sep-2019|
Arvind P Ganpule
Department of Urology, Muljibhai Patel Urological Hospital, Nadiad - 387 001, Gujarat
Source of Support: None, Conflict of Interest: None
Aim: To construct and validate a cost-effective indigenously made retrieval system (Modified Nadiad bag) in minimal access urology.
Method: The components of the modified Nadiad bag are a polyethylene roll, fishnet thread, and a 5 Fr ureteral catheter. The bag is indigenously made in our institute and used for organ retrieval after proper sterilization. The video recordings of entrapments and retrievals done over the past few months were reviewed. The procedures under review in which the bag was used were: Robot Assisted Radical Prostatectomy (50 cases), laparoscopic radical nephrectomy (50 cases), laparoscopic simple nephrectomy (50 cases) and laparoscopic adrenalectomy (18 cases). We also compared the retrieval time with experts and novices.
Results: The retrieval times, the organ size (largest dimension) and specimen weight were records for each case. Multivariate analysis of the data was done and we extrapolated the retrieval time with organ size, specimen weight and expertise of the surgeon. There was no significant difference among expert surgeons and novice surgeons with regards to retrieval times (p value = 0.29), with regards to organ size (p value = 0.83) and with regards to specimen weight (p value = 0.99).
Conclusion: Our design of retrieval system offers a cost-effective option which is easy to make, without the risk of tumor seeding and without the need for separate access sheath. It's a retrieval system which has proved its efficacy in laparoscopic as well as robotic procedures with no bearing on the expertise of the surgeon involved.
Keywords: Laparoscopic surgery, nadiad bag, retrieval system, robotic surgery, specimen, validation
|How to cite this article:|
Deshmukh CS, Ganpule AP, Islam MR, Sabnis RB, Desai MR. Laparoscopic and robotic specimen retrieval system (Modified Nadiad Bag): Validation and cost-effectiveness study model. J Min Access Surg 2019;15:305-10
|How to cite this URL:|
Deshmukh CS, Ganpule AP, Islam MR, Sabnis RB, Desai MR. Laparoscopic and robotic specimen retrieval system (Modified Nadiad Bag): Validation and cost-effectiveness study model. J Min Access Surg [serial online] 2019 [cited 2019 Sep 21];15:305-10. Available from: http://www.journalofmas.com/text.asp?2019/15/4/305/235398
| ¤ Introduction|| |
Specific retrieval forms an integral part of any ablative procedure. The key requisites for any successful entrapment of the specimen include expeditious retrieval and ensuring that the specimen is not mutilated which may hamper proper histological interpretation. There are also concerns regarding oncological principles such as spillage.
The currently available specimen retrieval bags cost in the range of 25–100 USD.,,, All laparoscopic surgeons have their indigenous ways of entrapping, the specimen using the available glove finger, etc. However, the drawbacks of all these indigenous entrapment materials include the inability to keep the 'mouth' of the entrapment device open. This increases the difficulty for entrapment besides adding to the operating time. This novel 'homemade' bags do not ensure retrieval without spillage.
We have devised a solution, a bag which is not only cost effective but is also as effective as the available devices. This bag was initially described as the Nadiad Bag. We performed a validation study in various urological procedures and assessed the safety, efficacy and cost-effectiveness. We also evaluate the efficacy by validating the entrapment time and retrieval time.
Aim of study
The aim of this study was to construct and validate a cost-effective indigenously made retrieval system in minimal access urology.
| ¤ Materials and Methods|| |
The components of the retrieval bag are a transparent plastic bag roll, fishnet thread and a 5 Fr ureteral catheter. The plastic bag rolls have a tubular construction throughout their length and three different sizes are available in the market [Figure 1]a. These bag rolls are unwound to the required length and cut according to the individual requirement [Figure 1]b. One end of the plastic bag roll is stitched completely and sealed which becomes the base of the bag [Figure 1]c. The corners of the other end (mouth) of the bag are cut up to 3 cm [Figure 1]d. The other corner of the mouth being cut in a similar manner [Figure 2]a. The plastic slip so created after cutting the corner is folded outwards on itself to create a tunnel [Figure 2]b. The folded edge is then stitched to create a tunnel [Figure 2]c. In a similar manner, the other slip of plastic is folded outwards on itself and stitched to create a second tunnel [Figure 2]d. The final bag so created has a sealed lower end and the mouth of the bag with two tunnels at the edges [Figure 3]a, [Figure 3]b. A loop is created at the end of the fishnet thread. The ureteric catheter is passed through this loop for ease of insertion of catheter in the tunnel [Figure 3]c. The folded tunnel accommodates the 5F ureteral catheter and fishnet thread [Figure 4]. Finally, the bag is folded compactly with rubber bands wound tightly around it and sealed in a plastic bag and sterilised using plasma steriliser before use.
|Figure 1: (a) Plastic Bag Rolls (b) Bag roll cut to a required length (c) Lower end of the bag stitched and closed (d) Corner of the open end being cut|
Click here to view
|Figure 2: (a) Opposite corner of the open end being cut (b) Slip of plastic being folded outward to create a tunnel (c) Folded edge is stitched to create a tunnel (d) Other slip also folded and stitched to create another tunnel|
Click here to view
|Figure 3: (a) Bag has a stitched bottom end and mouth with two tunnels. (b) Mouth of the bag with two tunnels. (c) 5 Fr ureteric catheters passed through fishnet loop|
Click here to view
|Figure 4: Tunnelling of the ureteric catheter along with the fishnet thread|
Click here to view
During the operative procedure, the bag introduction is performed through the 10-mm port with the help of an atraumatic grasper without any special introducer sheath. Two laparoscopic instruments are required to open, place the specimen within and close the bag [Figure 5]a, [Figure 5]b, [Figure 5]c. Once the specimen is secured in the bag, the ureteric catheter is pulled out with a laparoscopic grasper and the fishnet thread is pulled to close the open mouth of the bag. The bag can be retrieved though any of the ports by extending the incision if required. The bag is manufactured at Muljibhai Patel Urological Hospital research laboratory.
Ever since its induction in routine ablative laparoscopic procedures at our institute we have made two modifications as follows: (1) The use of fishnet thread instead of nylon sutures as fishnet thread is more compliant with plastic since it does not tear through the plastic bag when pulled which was the case with nylon thread and fishnet thread is more economical. (2) The lower end of the bag is stitched on a sewing machine which was previously sealed with an autoseal device. Earlier bags sealed with an autoseal device often showed dehiscence and opened when the specimen was placed.
It is available in three sizes. The sizes are as follows [Table 1]:
The key feature of the bag and the idea which makes this an innovation is the ureteric catheter. In all indigenous bags or the homemade bags, there is a shortcoming that the mouth of the bag cannot be kept open as the plastic keeps on folding on itself. We thought that we need to device a mechanism by which the mouth remains open. This can be kept open using a stiff material. The material can be a ureteric catheter.
Another major shortcoming associated with the commercially available retrieval systems is the cost ranging from $ 25 to 100,,, [Table 2].
|Table 2: Comparison of cost of currently available retrieval systems in the market|
Click here to view
The microcosting for our innovation is only over 1 USD [Table 3].
The present study was conducted at Muljibhai Patel Urological Hospital, Nadiad, India. The video recordings of entrapments and retrievals done over the past few months were reviewed. The procedures under review in which the bag was used were as follows: robot-assisted radical prostatectomy (50 cases), laparoscopic radical nephrectomy (50 cases), laparoscopic simple nephrectomy (50 cases) and laparoscopic adrenalectomy (18 cases). We also compared the retrieval time with experts and novices. The compatibility and permeability testing of the bag was performed using the innovative saline fill technique.
| ¤ Results|| |
The retrieval times for all the above-mentioned procedures were collected. The organ size (largest dimension) and specimen weight were also recorded for each case [Table 4].
The saline fill test was done in all models of the bag. It did not reveal any leak of saline.
Multivariate analysis of the data was performed, and we extrapolated the retrieval time with organ size, specimen weight and expertise of the surgeon [Table 5]. Statistical tests applied were Levene's test for equality of variances and t-test for equality of means. There was no statistically significant difference between expert surgeons and novice surgeons with regard to retrieval times (P = 0.29), with regard to organ size (P = 0.83) and with regard to specimen weight (P = 0.99).
| ¤ Discussion|| |
Even since the introduction of the laparoscopy as a modality in the realm of urology, it has gained widespread acceptance for various reasons such as shorter hospital stay, less postoperative pain and better cosmesis. However, one of the challenges faced in laparoscopic ablative surgeries is the extraction of the excised organ specimen.
Direct specimen retrieval attempted initially often required larger incisions and thus increasing the risk of an incisional hernia besides compromising cosmesis.
With the intent to tackle this challenge Clayman and associates designed the first organ bag (LapSac), especially for laparoscopic nephrectomy. Since then a number of versions of specimen retrieval systems have been developed and made commercially available for specimen retrieval intact as well as morcellation with specimen retrieval.
Currently, commercially available retrieval systems individually have their own advantages and disadvantages. The Pleatman sac by design has its mouth continuously open allowing the specimen placed inside to slip out while extraction. The Endocatch bag turns irretrievable once the purse string is activated closing and tearing of the plastic bag. The LapSac bag (Cook Urological) which is one of the most commonly used bags for morcellation of nephrectomy specimen often requires placement of two or three additional ports to manoeuvre specimen into the bag. This also further demands a learning curve to use this bag. The Lap Bag, Extraction Bag, Endobag, Endopouch and Endocatch are some retrieval systems which require an additional introducer sheath to deploy the bag. The Espiner bag has been proven to have poor impermeability.
The material used in manufacturing of the bags does play an important role in the success or failure of retrieval of an intact specimen. Singhvi and associates analysed the physical properties of the endoscopic retrieval systems and revealed that plastic systems could be delivered intact with little force when compared to fabric systems which tore in response to a significantly greater force. In addition, as plastic bags are nonporous, cells from neoplastic lesions are less like to leak thus reducing the risk of tumour seeding.
Above all, a major shortcoming associated with these retrieval systems is the cost. The cost factor does come into play in third world scenarios where the surgeons have to work with limited resources while maintaining the oncological principles and avoid compromising the quality of care. The cost of these retrieval systems ranges from $25 to 100; thus, increasing the cost of surgery to an extent discouraging the patients from opting for minimal access modality.
Keeping all these aspects in mind, we designed a retrieval bag as an economical option for specimen extraction. The ureteric catheter placed in the tunnel of the bag helps to keep the mouth of the bag open owing to its inherent stiffness with certain degree of malleability. This eases the entrapment of the specimen without the worry of edges of the bag folding inside. The bag being made of transparent plastic and sealed with an autoseal device ensures intact specimen retrieval without the risk of tumour seeding. The retrieval system does not require any separate introducer sheath for deployment or any additional ports for manoeuvring. Finally, the entire cost of making one bag turned out to be $1.20.
These observations do show the potential impact of this innovation can have in benefitting a large number of patients, especially from the underprivileged sections of the society. These are the patients for whom cost of a procedure is a factor that dictates whether he/she will opt for medical intervention or not.
Furthermore, the statistical analysis in this study clearly shows that the experience of the surgeon (expert or novice) did not have an impact on the retrieval time of the specimen. This finding suggests that the use of our retrieval system did not require the surgeons to undergo a steep learning curve.
| ¤ Conclusion|| |
Among the commercially available retrieval systems, our design of retrieval system offers a cost-effective option which is easy to make, without the risk of tumour seeding and the need for separate access sheath. It's a retrieval system which has proved its efficacy in laparoscopic as well as robotic procedures with no bearing on the expertise of the surgeon involved. Thus Modified Nadiad Bag can be an option worth considering in circumstances where the cost of surgery matters as much as the quality of surgery.
Limitations of the study
- Any device or technique of procedure, when introduced into clinical practice is pitched against a gold standard device or technique. Similarly, there is a need for the nadiad bag to be compared to a retrieval system already available in the market to test its efficiency
- We acknowledge that the study was a retrospective one. A two-arm prospective study with an equal number of participants in expert group and novice group would be a more scientifically ideal way of knowing if the expertise of the surgeon does have a bearing in using this retrieval system or not.
Financial support and sponsorship
Conflicts of interest
There are no conflicts of interest.
| ¤ References|| |
Ganpule AP, Gotov E, Mishra S, Muthu V, Sabnis R, Desai M, et al.
Novel cost-effective specimen retrieval bag in laparoscopy: Nadiad bag. Urology 2010;75:1213-6.
Clayman RV, Kavoussi LR, Long SR, Dierks SM, Meretyk S, Soper NJ. Laparoscopic nephrectomy: Initial report of pelviscopic organ ablation in the pig. J Endourol 1990;4:247-52.
Rassweiler J, Stock C, Frede T, Seemann O, Alken P. Organ retrieval systems for endoscopic nephrectomy: A comparative study. J Endourol 1998;12:325-33.
Singhvi SK, Allan W, Williams ED, Small PK. Assessment of the physical properties of endoscopic retrieval systems. Br J Surg 2002;89:1183-7.
[Figure 1], [Figure 2], [Figure 3], [Figure 4], [Figure 5]
[Table 1], [Table 2], [Table 3], [Table 4], [Table 5]